Hi,

I am currently using Pololu AltIMU-10 v4. However, I am confused by the datasheet provided.

For example, for accelerometer, we usually take the raw data and divide it by 65,535 (16 bit data) and then multiply by the linear acceleration measurement range. However, there is another way of conversion is done by directly multiplying with linear acceleration sensitivity.

For measurement range of +/- 2, 4, 6 and 8, the result obtained through the calculation above and linear sensitivity multiplication are the same.

4 / 65535 = 0.061 mg/LSB

8 / 65535 = 0.122 mg/LSB

12 / 65535 = 0.183 mg/LSB

16 / 65535 = 0.244 mg/LSB

They are the same for the four measurement range above. However, for +/- 16 g, the result is different from linear acceleration sensitivity as shown in datasheet (0.732 mg/LSB).

32 / 65535 = 0.488 mg/LSB

In this case, which method should I use? Should I multiply it by sensitivity directly or follow the calculation above? In fact, I had found that this calculation method is done by a lot of people but I cannot find it in the st datasheet.

The same issue occur with the magnetometer on the same chip (LSM303D) and the gyroscope chip on the same board (L3GD20H).

I hope that someone can clarify this issue with me.

Thank you.

Regards,

qzai09