My LSM303D is having an offset (about -2.8 g) in x. The accelerometer has returned valid values until now. Has someone encountered anything similar? What could be the cause?
Is it possible to re-calibrate the LSM303D using its features? I could do some compensation after reading the data, but wonder if the LSM303D could be “reset” somehow. I found a “Reboot memory content” in the data sheet, I don’t know if it would help me.
I have attached a figure of the sensor readings, while the sensor is sitting flat on table (note the correct readings in y and z).
The LSM303D is integrated on the AltIMU-10 v4.
Thanks in advance!